# SOCCOM Additional Information - 18-Mar-2018 05:04:29
7567,5904182, 03 Feb 2016: change notation to MPD. Battery has died while float was under ice, was weakening before it went under earlier in 2015
9092,5904185, 22 Jan 2017: Additional CTD calibration cast from P18, EXPOCODE=33RO20161119, Station=195, Cast=1
9031,5904396, 17 Jan 2017: appears to have died at half of battery capacity - probably blew 2 fuses
9018,5904186, Apr 2016: Problem with conductivity, which is also used to calculate oxygen
9095,5904188, 7 Jun 2016: grounded on 800m deep ridge south of Heezen Fracture Zone after 113 profiles
06 Jul 2016: change notation to MPD.
29 Nov 2016: change notation to active: no longer stuck in mud, profile 130
06 Jul 2016: change notation to MPD.
06 Mar 2018: Batteries Dead
9254,5904395, 2017 Oct 3 - will be recovered, battery prematurely low. Probably recovered about Oct 17
9254,5904395, 2017 Oct 31 - was recovered by this date
0037,5904475, 18 Jan 2016: float stuck at surface starting 26 Dec 2015 profile 69 - air/oil separator valve stuck
19 July 2016: began profiling again on 9 July 2016 profile 89
03 Aug 2016: float is responding again, removed the stuck surface designation
9313,5904474, 2 Aug 2016: Now too far into subtropics for standard eMLR algorithm to work properly
13 Jun 2017: pH bad starting cycle 108
0508,5904476, 17 Jan 2015: pH sensor failed on deployment.
07 Feb 2016: induced into under-ice mode for testing, following profile 42 on January 23, 2016, until end of February, 2016.
19 Apr 2016: Declared MPD after missing 42 profiles. Problem with maintaining buoyancy, similar failure mode as several other Seabird Navis floats (0509, 0511) from the same deployment group.
12 Jul 2016: Potential that float grounded at Marion Island, but most likely buoyancy failure as noted in April.
0509,5904477, 2 June 2015: last good profile before ice cover, did not emerge, declared MPD fall 2015
0511,5904478, 17 Jan 2015: Stuck at surface after 3 good profiles, buoyancy problem
13 June 2015: UW data acquisition terminated
9094,5904471, 07 Feb 2016: pH flag: removed comments from float status page, put comments in additional information file
9275,5904472, 24 Jan 2018: pH bad after cycle 108
9260,5904473, 17 Jan 2017: sending partial files, but not completing the transmissions. Pneumatic system has failed, bladder fails to inflate, so antenna remains submerged too long to communicate.
9260,5904473, 27 Feb 2017 Lost Buoyancy. Can not stay at surface long enough to transmit.
8514,5904470, 2 Aug 2016: pH sensor becoming bad at profile 54
8514,5904470, 10 Oct 2017: Dead after 96 profiles, last profile on 2017-09-10.This was a float that was used in the initial pH sensor tests off Hawaii and was refurbished for deployment in SOCCOM.
9666,5904662, 18 Jan 2016: salinity problems, not likely salvageable (Swift)
12 July 2016: bad salinity and other properties below 1000 on profiles 2-7, 11-13, 14, 18; salinity spiking on many of these and also 10, 13, 15
9652,5904660, 07 Feb 2016: keep watch on oxygen; sensor cleaned prior to deployment
9652,5904660, 22 Nov 2017: Additional CTD calibration cast from UK COMICS program (and ORCHESTRA), about 14 hours behind the float surface time, and near the surface location.
9657,5904659, 07 Feb 2016: keep watch on oxygen; sensor cleaned prior to deployment
9645,5904676, 7 June 2016: 9 bad nitrate profiles, can deal with in calibration
0506,5904670, 07 Feb 2016: pH flag: removed comments from float status page, put comments in additional information file
0506,5904670, 13 Jan 2017: Dead CTD
0506,5904670, 10 Aug 2017: The root-cause of the salinity failure on navis 0506 is that the float experienced flooding after profile 035. This diagnosis is pretty conclusive as it involves many interlocking bits of evidence & data. The symptoms are also many ranging from buoyancy issues, faulty valve operations, salinity, malfunction of the pneumatic system, poor GPS and telemetry, malfunction of the PPI that services the suna, and on and on.
9600,5904688, 03 May 2016: 6 profiles completed. pH sensor is beginning to become intermittent, in same failure mode as other pH sensors
28 June 2016: pH sensor continuing with problems
9631,5904677, 27 Apr 2017: Profiles starting at 33, on Jan. 14, 2017, have been in water around 1000 m on the Campbell Plateau.
0568,5904767, 5 May 2016: pH first profiles shakey
0565,5904672, 09 May 2016: pH sensor not working from start, likely due to leak between back of ISFET and seawater
09 May 2016: Optical sensor has chlorophyll fluorescence and backscatter at 530 and 700 nm instead of chlorophyll and CDOM fluorosence plus BB at 700 nm
9761,5904764, 12 Jul 2016: CApex (carbon fiber hull Apex) float over-ballasted and probably descended below its crush depth on the 2nd profile and imploded
0571,5904673, 17 Jan 2017: pH sensor failed from outset, data are bad
0566,5904766, 16 May 2016: First pH profile with insufficiently conditioned sensor (SBE-built pH sensor), should improve
9660,5904761, 17 May 2016: bottom of float hit side of ship on deployment, no external sensors were involved.
9762,5904765, 12 Jul 2016: Profiles cleaned up and online. Profile 2 has salinity and FLBB backscatter QC problem, possibly something attached to float and then fell off
9659,5904843, 9659,5904843, 12 Apr 2017: O2 sensor bad; pH still looks good; NO3 flagged good even though O2 (used in qc process) is bad
9642,5904685, 27 Dec 2017: pH failed at cycle 37
12382,5904841, 12 Jan 2017: Nitrate sensor light throughput is ~10% of calibration. Cy2 will help determine cause.
Failed NO3 and pH, O2 gain assessed
9750,5904845, 23 Jan 2017: Gouges with paint scrapes on hull prior to deployment, repaired with light enamel and deployed
8501,5905075, 23 Jan 2017: Gouges with paint scrapes on hull prior to deployment, repaired with light enamel and deployed
12559,5904855, 29 Jan 2017: Additional CTD calibration cast from P18, EXPOCODE=33RO20161119, Station=212, Cast=1
12549,5904859, 13 Oct 2017: pH failed at Cy 29
12551,5904858, 10 Jan 2016: optode failed on first cycle: unknown cause
12541,5904860, 29 Dec 2017: pH failed at cycle 29
0692,5905073, 17 Jan 2017: pH sensor failed from outset, data are bad
12361,5905100, 18 Apr 2017: pH sensor is bad from outset.
12363,5905102, 30 May 2017: pH sensor went bad on Cy 5
12369,5905106, 26 Sept 2017: bad Aanderaa optode oxygen sensor on first profile, most likely a manufacturing defect
12729,5905133, 19 Sep 2017: pH failure upon deployment: As I've noted before, we see 3 different failure mechanisms based on diagnostic data that comes back. The fixes we've implemented might cure the first two, which are most common. The pH on float 12729 failed with the 3rd, more rare mechanism. Turns out there was some engineering data in the dock tests that indicated this might occur, but no one noticed. We're reviewing all the dock test data for other floats to verify they don't have this issue. When we were producing this batch of pH sensors, we had to scrap half a dozen or so with similar problems. It was due to a production short cut that the technicians put in place to speed things up.
12729,5905133, 26 Sep 2017: bad pH sensor on first profile, most likely an assembly defect
12745,5905132, 10 Oct 2017: pH sensor problems
12733,5905131, 8 Oct 2017 - box hit the water a bit hard on deployment
0690,5905071, 16 Jan 2018 - has bad pH sensor
0693,5905074, 12 Jan 2018: Navis float CTD failed, which means complete failure of float after the second profile. Failure occurred during parking phase. This float also had excess grease around the end cap prior to deployment and caused concern on part of I. Rosso, but deployment was OKed by D. Swift.
12757,5905366, 06 Feb 2018: pH and nitrate sensors are bad. float got knowcke around on the Agulhas
0688,5904846, 16 Jan 2018: pH sensor failed on first profile